Many SIGLENT instruments feature options that can be added by using an activation code. To generate the activation code: – Purchase the option from your favorite SIGLENT office or distributor. A full list of available options are located at the bottom of the specific product webpage. – You will receive an Option Card PDF document … Continued
The X series of SIGLENT SDG arbitrary generators, including the 1000X, 2000X, and 6000X models, offer a large array of adjustments and features that can make common tasks quite simple. One common application is to add a certain amount of noise to a standard waveform and then adjust the characteristics of the noise level. This … Continued
The Universal Serial Bus (USB) is very popular and has added a tremendous level of convenience and flexibility to computer and peripheral controls, including test and measurement instrumentation. As with all technology, it isn’t always sunshine and blue skies. Here are some common troubleshooting steps for USB connection issues: – Is the instrument plugged in … Continued
The installation of SIGLENT EasySpectrum and other “Easy” software is typically uneventful. But, there may be an occasional issue. Here are some steps that may help clear up any issues that you may encounter: Here is an example of a mis-rendered window view of EasySpectrum: Here are some steps that may help fix the issue: … Continued
The SIGLENT SDG6000X series of Arbitrary Waveform Generators (AWG) have the ability to source IQ modulated waveforms with the option SDG6000X-IQ. To activate the IQ capabilities: Purchase the SDG6000X-IQ option from your nearest SIGLENT Authorized distributor or direct office. Follow the directions to create an activation code. 2. Power on the generator and press the Waveforms … Continued
The SDGs require UTF8 formatting for the words used to build an arbitrary waveform. The code may report “UnicodeDecodeError: ‘ascii’ codec can’t decode byte 0x80 in position 77: ordinal not in range(128)” This indicates that the arbitrary waveform data is coded as “ascii” which is not the expected data format “utf8”. Here is the proper … Continued
All SIGLENT instruments (except for the SHS1000 and SHS800 hand-held oscilloscopes) support SCPI remote control commands. For more information, download the specific instrument programming guide located on the product page.
Excerpt from the ANSI-ASQ National Accreditation Board: http://anab.org/news/newsletter/winter-2014/nist-traceability-statements NIST Traceability Statements Many calibration customers request on their accredited calibration certificates a statement that the measurement is traceable to the National Institute of Standards and Technology (NIST). They want assurance that the standards used for the calibration are ultimately traceable to the International System of Units … Continued