SIGLENT instruments with USB ports can supply 500mA steady current at 5 V.
The SIGLENT SPD series of power supplies have only positive voltage and current setpoint values. That is, you can select voltages or currents that are from 0 to the max value (0-32 V, for voltage.. as an example for the SDP3000X/X-E). Here are the suggested wiring configurations to achieve both positive and negative potentials at the … Continued
Information regarding product compliance is available on the datasheet for each product. Contact your local Siglent office for more details if you have trouble finding what you need. You can also write firstname.lastname@example.org
The SIGLENT SPD series of power supplies offer constant current mode. This setting limits the current output of the specified channel and can be used as a constant current source and to protect your device-under-test (DUT). In this tip, we show the typical performance of the SPD as it switches between constant voltage (CV) and … Continued
The SPD series of power supplies feature remote programming capabilities and can be controlled via USB or LAN connection. In this note, we will use Python and PyVISA to create a program that will connect to an SPD, enable the output, delay, and then disable the output. There are two areas that differ from some … Continued
Test and measurement equipment is exempt from FCC emissions limits per CFR 47, Part 15, para 15.103(c): https://www.gpo.gov/fdsys/pkg/CFR-2002-title47-vol1/pdf/CFR-2002-title47-vol1-sec15-103.pdf Specifically: “(c) A digital device used exclusively as industrial, commercial, or medical test equipment.”
The Universal Serial Bus (USB) is very popular and has added a tremendous level of convenience and flexibility to computer and peripheral controls, including test and measurement instrumentation. As with all technology, it isn’t always sunshine and blue skies. Here are some common troubleshooting steps for USB connection issues: – Is the instrument plugged in … Continued
All SIGLENT instruments (except for the SHS1000 and SHS800 hand-held oscilloscopes) support SCPI remote control commands. For more information, download the specific instrument programming guide located on the product page.
Linear DC regulated power supplies (“linear supplies”) alter the output voltage to the required value through the power frequency (50/60 Hz alternating current) voltage transformer. After being lowered (or raised) to the appropriate AC voltage value, the waveform is rectified, filtered, and stabilized by a voltage regulation circuit, and is then output as a regulated … Continued
Excerpt from the ANSI-ASQ National Accreditation Board: http://anab.org/news/newsletter/winter-2014/nist-traceability-statements NIST Traceability Statements Many calibration customers request on their accredited calibration certificates a statement that the measurement is traceable to the National Institute of Standards and Technology (NIST). They want assurance that the standards used for the calibration are ultimately traceable to the International System of Units … Continued