SIGLENT instruments with USB ports can supply 500mA steady current at 5 V.
The term “div” for most SIGLENT specifications is equal to the Vertical Scale value (Commonly shown in units of Volts/div). Take the specification for the DC Gain Accuracy of a SIGLENT SDS1052DL+ For ranges ≤ 100 mV/div, the DC Gain Accuracy is given as ± [3% × (|reading|+|offset|) +1% × |offset| +0.2 * div+2 mV] If: Reading = 0.5 … Continued
Information regarding product compliance is available on the datasheet for each product. Contact your local Siglent office for more details if you have trouble finding what you need. You can also write email@example.com
Test and measurement equipment is exempt from FCC emissions limits per CFR 47, Part 15, para 15.103(c): https://www.gpo.gov/fdsys/pkg/CFR-2002-title47-vol1/pdf/CFR-2002-title47-vol1-sec15-103.pdf Specifically: “(c) A digital device used exclusively as industrial, commercial, or medical test equipment.”
The Universal Serial Bus (USB) is very popular and has added a tremendous level of convenience and flexibility to computer and peripheral controls, including test and measurement instrumentation. As with all technology, it isn’t always sunshine and blue skies. Here are some common troubleshooting steps for USB connection issues: – Is the instrument plugged in … Continued
The installation of SIGLENT EasySpectrum and other “Easy” software is typically uneventful. But, there may be an occasional issue. Here are some steps that may help clear up any issues that you may encounter: Here is an example of a mis-rendered window view of EasySpectrum: Here are some steps that may help fix the issue: … Continued
Here is a list of probes currently supplied with each oscilloscope: Click here to view the Product Probe List
The SDS series of SIGLENT Digital Oscilloscopes have a number of save options that include copies of the displayed image (Bitmap, JPG, and PNG are common image types) as well as instrument configurations (Setups) and data (CSV). Here are brief instructions on how to save files to external USB memory devices. Insert a USB memory … Continued
All SIGLENT instruments (except for the SHS1000 and SHS800 hand-held oscilloscopes) support SCPI remote control commands. For more information, download the specific instrument programming guide located on the product page.
In some extreme hardware errors, the oscilloscope may become unresponsive or “hang” during the boot sequence. If your oscilloscope is frozen, will not boot, or appears to be unresponsive, you can attempt a hardware reset by following these instructions: 1. Disconnect all USB, LAN, and input cables. Leave the power cord connected. 2. Power cycle … Continued