The most common reason that the SPD3303-C is stuck in constant current mode (CC.. indicated by a red LED between the channel outputs) is an incorrect voltage setting on the inputs. Here is a picture of the voltage settings at 220 V, but with 120 V input power: Note that with no load attached, the … Continued
The Universal Serial Bus (USB) is very popular and has added a tremendous level of convenience and flexibility to computer and peripheral controls, including test and measurement instrumentation. As with all technology, it isn’t always sunshine and blue skies. Here are some common troubleshooting steps for USB connection issues: – Is the instrument plugged in … Continued
The installation of SIGLENT EasySpectrum and other “Easy” software is typically uneventful. But, there may be an occasional issue. Here are some steps that may help clear up any issues that you may encounter: Here is an example of a mis-rendered window view of EasySpectrum: Here are some steps that may help fix the issue: … Continued
All SIGLENT instruments (except for the SHS1000 and SHS800 hand-held oscilloscopes) support SCPI remote control commands. For more information, download the specific instrument programming guide located on the product page.
The SIGLENT SPD3000 series of power supplies can be upgraded remotely using a computer, USB cable, and SIGLENT EasyPower software. 1. Download EasyPower from the firmware & software tab of the SPD Product Page 2. Download the latest firmware from the firmware and software tab of the SPD Product Page. Unzip the file using a … Continued
Linear DC regulated power supplies (“linear supplies”) alter the output voltage to the required value through the power frequency (50/60 Hz alternating current) voltage transformer. After being lowered (or raised) to the appropriate AC voltage value, the waveform is rectified, filtered, and stabilized by a voltage regulation circuit, and is then output as a regulated … Continued
Excerpt from the ANSI-ASQ National Accreditation Board: http://anab.org/news/newsletter/winter-2014/nist-traceability-statements NIST Traceability Statements Many calibration customers request on their accredited calibration certificates a statement that the measurement is traceable to the National Institute of Standards and Technology (NIST). They want assurance that the standards used for the calibration are ultimately traceable to the International System of Units … Continued
What instruments accept SCPI commands for remote control? All SIGLENT instruments (except for the SHS1000 and SHS800 hand-held oscilloscopes) support SCPI remote control commands. For more information, download the specific instrument programming guide located on the product page.